Read data from instrument, and format as text
A = fscanf(obj)
A = fscanf(obj,'format')
A = fscanf(obj,'format',size)
[A,count] = fscanf(...)
[A,count,msg] = fscanf(...)
[A,count,msg,datagramaddress] = fscanf(obj,...)
[A,count,msg,datagramaddress,datagramport] = fscanf(obj,...)
An interface object.
C language conversion specification.
The number of values to read.
Data read from the instrument and formatted as text.
The number of values read.
A message indicating if the read operation was unsuccessful.
The address of the datagram sender.
The port of the datagram sender.
format is a C language conversion specification. Conversion specifications involve the % character and the conversion characters d, i, o, u, x, X, f, e, E, g, G, c, and s. Refer to the sscanf file I/O format specifications or a C manual for more information.
Read at most n values into a column vector.
Read at most m–by–n values filling an m–by–n matrix in column order.
size cannot be inf, and an error is returned if the specified number of values cannot be stored in the input buffer. If size is not of the form [m,n], and a character conversion is specified, then A is returned as a row vector. You specify the size, in bytes, of the input buffer with the InputBufferSize property. An ASCII value is one byte.
If obj is a UDP object and DatagramTerminateMode is off, the size value is honored. If size is less than the length of the datagram, only size values are read. If size is greater than the length of the datagram, a warning is issued stating that a complete datagram was read before size values was reached.
Create the serial port object s on a Windows® machine and connect s to a Tektronix® TDS 210 oscilloscope, which is displaying a sine wave.
s = serial('COM1'); fopen(s)
Use the fprintf function to configure the scope to measure the peak-to-peak voltage of the sine wave, return the measurement type, and return the peak-to-peak voltage.
fprintf(s,'MEASUREMENT:IMMED:TYPE PK2PK') fprintf(s,'MEASUREMENT:IMMED:TYPE?') fprintf(s,'MEASUREMENT:IMMED:VAL?')
Because the default value for the ReadAsyncMode property is continuous, data associated with the two query commands is automatically returned to the input buffer.
s.BytesAvailable ans = 13
Use fscanf to read the measurement type. The operation will complete when the first terminator is read.
meas = fscanf(s) meas = PK2PK
Use fscanf to read the peak-to-peak voltage as a floating-point number, and exclude the terminator.
pk2pk = fscanf(s,'%e',6) pk2pk = 2.0200
Disconnect s from the scope, and remove s from memory and the workspace.
fclose(s) delete(s) clear s
Before you can read data from the instrument, it must be connected to obj with the fopen function. A connected interface object has a Status property value of open. An error is returned if you attempt to perform a read operation while obj is not connected to the instrument.
If msg is not included as an output argument and the read operation was not successful, then a warning message is returned to the command line.
The ValuesReceived property value is increased by the number of values read — including the terminator — each time fscanf is issued.
A read operation with fscanf blocks access to the MATLAB® command line until
The terminator is read. For serial port, TCPIP, UDP, and VISA-serial objects, the terminator is given by the Terminator property. If Terminator is empty, fscanf will complete execution and return control when another criterion is met. For UDP objects, DatagramTerminateMode must be off.
For all other interface objects, the terminator is given by the EOSCharCode property.
The time specified by the Timeout property passes.
The number of values specified by size is read. For UDP objects, DatagramTerminateMode must be off.
A datagram is received (for UDP objects only when DatagramTerminateMode is on).
The input buffer is filled.
The EOI line is asserted (GPIB and VXI instruments only).
The EOSCharCode property value is recognized only when the EOSMode property is configured to read or read&write. For example, if EOSMode is configured to read and EOSCharCode is configured to LF, then one of the ways that the read operation terminates is when the line feed character is received.
If EOSMode is none or write, then there is no terminator defined for read operations. In this case, fscanf will complete execution and return control to the command when another criterion, such as a timeout, is met.