CIO-DAC16 Analog Output (D/A) block (not recommended)
Simulink® Real-Time™ Library of Drivers No Longer Recommended for Use
I/O Module Output
Block Input Data Type
Enter numbers between 1 and 16. This driver allows the selection of individual D/A channels in arbitrary order. The number of elements defines the number of D/A channels used. For example, to use the first and second analog output (D/A) channels, enter
Number the channels beginning with 1 even if this board manufacturer starts numbering the channels with 0.
Enter a range code for each of the channels in the channel vector. The range vector must be the same length as the channel vector. This board allows the range of each channel to be different.
The following table is a list of the ranges for this driver and the corresponding range codes.
Input Range (V)
-10 to +10
-5 to +5
-2.5 to +2.5
0 to +10
0 to +5
0 to +2.5
For example, if the first channel is -10 to +10 volts and the second channel is 0 to +5 volts, enter
The range settings must correspond to the DIP switch settings on the board.
The reset vector controls the behavior of the channel at model
termination. Enter a scalar or a vector that is the same length as
the channel vector. If you specify a scalar value, that setting is
replicated over the channel vector. If you specify a value of
the corresponding channel is reset to the value specified in the initial
value vector. If you specify a value of
channel remains at the last value attained while the model was running.
The initial value vector contains the initial voltage values for the output channels. Enter a scalar or a vector that is the same length as the channel vector. If you specify a scalar value, that value is replicated as the initial value over the channel vector. The channels are set to the initial values between the time the model is downloaded and the time it is started.
Enter the base sample time or a multiple of the base sample
-1 means sample time is inherited).
Enter the base address of the board. It is important that this entry corresponds to the DIP switch settings on the board. For example, if the base address is 300 (hexadecimal), enter
The jumpers by the range DIP switches on the board must be in the XFER position. The Wait-State jumper has to be in the off position.