PC104-DAS16JR/16 Analog Input block
Simulink® Real-Time™ Library for Measurement Computing™
I/O Module Input
Block Output Data Type
If you select 16 channels (Input coupling parameter
16 single-ended channels) enter a number
between 1 and 16 to select the number of A/D channels used. If you
select eight channels (Input coupling parameter
8 differential channels) enter a number
between 1 and 8. This driver does not allow the selection of individual
channels or to mix single-ended and differential inputs.
Number the lines beginning with 1 even if this board manufacturer numbers the lines beginning with 0.
From the list, choose either
+-10V (-10 volts
to +10 volts),
0-1.25V. This driver does not allow the selection
of different range for each channel.
From the list, select one from the following list of input modes:
Single-ended channels (16 channels)
Differential channels (8 channels)
This choice must correspond to the MUX-switch setting on the board.
Enter the base sample time or a multiple of the base sample
-1 means sample time is inherited).
Enter the base address of the board. This entry must correspond
to the DIP switch setting on the board. For example, if the DIP switch
setting is 300 (hexadecimal), enter