PCIM-DAS1602/16 Analog Input block
Simulink® Real-Time™ Library for Measurement Computing™
I/O Module Input
Block Output Data Type
If you select 16 channels (Input coupling parameter
16 single-ended channels) enter a number
between 1 and 16 to select the number of A/D channels used. If you
select eight channels (Input coupling parameter
8 differential channels) enter a number
between 1 and 8. This driver does not allow the selection of individual
channels or to mix single-ended and differential inputs.
Number the lines beginning with 1 even if this board manufacturer numbers the lines beginning with 0.
From the list, choose either
+-10V (-10 volts
to +10 volts),
0-1.25V. This driver does not allow the selection
of a different range for each channel.
If a bipolar range is used, the bipolar switch on the board must be in the bipolar position. If a unipolar range is used, the bipolar switch must be in the unipolar position.
From the list, select one from the following list of input modes:
Single-ended channels (16 channels)
Differential channels (8 channels)
This choice must correspond to the MUX-switch setting on the board.
Enter the base sample time or a multiple of the base sample
-1 means sample time is inherited).
If only one board of this type is in the target computer, enter
locate the board.
If two or more boards of this type are in the target computer,
enter the bus number and the PCI slot number of the board associated
with this driver block. Use the format
To determine the bus number and the PCI slot number, type:
tg = slrt; getPCIInfo(tg, 'installed')