MATLAB and Simulink resources for Arduino, LEGO, and Raspberry Pi
Learn moreOpportunities for recent engineering grads.
Apply TodayNew to MATLAB?
MATLAB Central > MATLAB Newsreader > Problem reading data from Tektronix oscilloscope 

Subject: Problem reading data from Tektronix oscilloscope From: james Date: 8 Dec, 2009 20:14:02 Message: 1 of 6 
Dear all, 
Subject: Problem reading data from Tektronix oscilloscope From: Trent Jarvi Date: 8 Dec, 2009 23:35:04 Message: 2 of 6 

Subject: Problem reading data from Tektronix oscilloscope From: mario Farrugia Date: 29 Jan, 2010 16:31:22 Message: 3 of 6 
"Trent Jarvi" <tjarvi@mathworks.com> wrote in message <hfmnr8$5fs$1@fred.mathworks.com>... 
Subject: Problem reading data from Tektronix oscilloscope From: james Date: 1 Feb, 2010 09:53:02 Message: 4 of 6 
"mario Farrugia" <eexmf3@nottingham.ac.uk> wrote in message <hjv2gq$f9c$1@fred.mathworks.com>... 
Subject: Problem reading data from Tektronix oscilloscope From: Bo Date: 6 Nov, 2012 21:52:08 Message: 5 of 6 
I met this problem too. Now I have solved this problem. Actually there are 2 properties we need to modify. The first one is 'Endingpoint', which belongs to waveform properties. The default number is 10000, we could set that to be the same as Recordlength by editing the driver file *.mdd. The other one is input buffer size. That one could be configured by the set command. It should be at least 2 times larger than the recordlength. After that I am able to get a waveform which contains 10M points. 
Subject: Problem reading data from Tektronix oscilloscope From: Clayton Chu Date: 14 Dec, 2012 01:54:09 Message: 6 of 6 
"Bo" wrote in message <k7c0q8$d5n$1@newscl01ah.mathworks.com>... 
A tag is like a keyword or category label associated with each thread. Tags make it easier for you to find threads of interest.
Anyone can tag a thread. Tags are public and visible to everyone.