Note: You can use hardware from this vendor only with 32-bit MATLAB®. You can install a 32-bit MATLAB on 64-bit Windows®. For more information, see this technical bulletin.
For boards that do not have a channel gain list, an
error occurs at
start if all the channel input
ranges are not the same or the channel scan order is not contiguous.
However, if the
ClockSource property value is set
software, this rule does not apply.
You should configure the
setverify function just before starting
the hardware. Note that the
SampleRate value is
dependent upon the number of channels added to the device object.
For boards that do not support continuous background
transfer mode (i.e., the board does not have hardware clocking), the
ClockSource property value is
When running at a sampling rate of 5000 Hz or higher
and with a
TransferMode property value of
there may be a considerable decline in system performance.
Most boards do not support simultaneous input and output. However, if software clocking is used, then this limitation does not apply.
To use hardware digital triggers with the PCI-DAS4020/12 board, you must first configure the appropriate trigger mode with InstaCal.
Expansion boards are not supported. This includes the CIO-EXP family of products.
MEGA-FIFO hardware is not supported.